Citation Information :
Khan AA, Gupta PK, Baranwal AK, Jayashree M, Sahoo T. Comparison of Blood Pressure Measurements by Currently Available Multiparameter Monitors and Mercury Column Sphygmomanometer in Patients Admitted in Pediatric Intensive Care Unit. Indian J Crit Care Med 2023; 27 (3):212-221.
Background: The multiparameter monitor (MPM) is replacing mercury column sphygmomanometers (MCS) in acute care settings. However, data on the former's accuracy in critically ill children are scarce and mostly extrapolated from adults. We compared non-invasive blood pressure (NIBP) measurements by MPMs with MCS in pediatric intensive care unit (PICU).
Patients: Adequately sedated and hemodynamically stabilized children (age, 1–144 months) were prospectively enrolled.
Materials and methods: Three NIBP measurements were obtained from MCS (Diamond®, India) and MPM (Intellivue MX800® or Ultraview SL®) in rapid succession in the upper limb resting in supine position. Respective three measurements were averaged to obtain a paired set of NIBP readings, one each from MCS and MPM. Such readings were obtained thrice a day. NIBP readings were then compared, and agreement was assessed.
Results: From 39 children [median age (IQR), 30 (10–72) months], 1,690 sets of NIBP readings were obtained. A-third of readings were from infants and children >96 months, while 383 (22.6%) readings were from patients on inotropes. Multiparameter monitors gave significantly higher NIBP readings compared to MCS [median systolic blood pressure (SBP), 6.5 (6.4–6.7 mm Hg); diastolic blood pressure (DBP), 4.5 (4.3–4.6 mm Hg); mean arterial pressure (MAP), 5.3 (5.1–5.4 mm Hg); p < 0.05]. It was consistent across age, gender, and critical care characteristics. Multiparameter monitors overestimated SBP in 80% of readings beyond the maximal clinically acceptable difference (MCAD).
Conclusions: Non-invasive blood pressure readings from MCS and MPMs are not interchangeable; SBP was 6–7 mm Hg higher with the latter. Overestimation beyond MCAD was overwhelming. Caution is required while classifying systolic hypotension with MPMs. Confirmation with auscultatory methods is advisable. More studies are required to evaluate currently available MPMs in different pediatric age groups.
Stergiou GS, Boubouchairopoulou N, Kollias A. Accuracy of automated blood pressure measurement in children: Evidence, issues, and perspectives. Hypertension 2017;69(6):1000–1006. DOI: 10.1161/HYPERTENSIONAHA.116.08553.
Stergiou GS, Alpert B, Mieke S, Asmar R, Atkins N, Eckert S, et al. A universal standard for the validation of blood pressure measuring devices: Association for the advancement of medical instrumentation/European society of hypertension/International organization for standardization (AAMI/ESH/ISO) collaboration statement. J Hypertens 2018;36(3):472–478. DOI: 10.1097/HJH.0000000000001634.
Wong S-N, Tz Sung RY, Leung LC-K. Validation of three oscillometric blood pressure devices against auscultatory mercury sphygmomanometer in children. Blood Press Monit 2006;11(5):281–291. DOI: 10.1097/01.mbp.0000209082.09623.b4.
Pickering TG, Hall JE, Appel LJ, Falkner BE, Graves J, Hill MN, et al. Recommendations for blood pressure measurement in humans and experimental animals: part 1: blood pressure measurement in humans: a statement for professionals from the subcommittee of professional and public education of the American Heart Association council on high blood pressure research. Circulation 2005;111(5):697–716. DOI: 10.1161/01.CIR.0000154900.76284.F6.
Flynn JT, Pierce CB, Miller ER 3rd, Charleston J, Samuels JA, Kupferman J, et al. Reliability of resting blood pressure measurement and classification using an oscillometric device in children with chronic kidney disease. J Pediatr 2012;160(3):434–440. DOI: 10.1016/j.jpeds.2011.08.071.
Lurbe E, Agabiti-Rosei E, Cruickshank JK, Dominiczak A, Erdine S, Hirth A, et al. 2016 European society of hypertension guidelines for the management of high blood pressure in children and adolescents. J Hypertens 2016;34(10):1887–1920. DOI: 10.1097/HJH.0000000000001039.
Flynn JT, Kaelber DC, Baker-Smith CM, Blowey D, Carroll AE, Daniels SR, et al. Clinical practice guideline for screening and management of high blood pressure in children and adolescents. Pediatrics 2017;140(3):e20171904. DOI: 10.1542/peds.2017-1904.
Weiss SL, Peters MJ, Alhazzani W, Agus MSD, Flori HR, Inwald DP, et al. Surviving sepsis campaign international guidelines for the management of septic shock and sepsis-associated organ dysfunction in children. Pediatr Crit Care Med 2020;21(2):e52–e106. DOI: 10.1097/PCC.0000000000002198.
Fleegler E, Kleinman M. Pediatric advanced life support (PALS). In UpToDate (Eds. Wiley JF, Torrey SB), available from: https://www.uptodate.com/contents/pediatric-advanced-life-support-pals (Accessed on: 6 June 2021).
Ribezzo S, Spina E, Di Bartolomeo S, Sanson G. Noninvasive techniques for blood pressure measurement are not a reliable alternative to direct measurement: A randomized crossover trial in ICU. ScientificWorldJournal 2014;2014:353628. DOI: 10.1155/2014/353628.
Bur A, Hirschl MM, Herkner H, Oschatz E, Kofler J, Woisetschläger C, et al. Accuracy of oscillometric blood pressure measurement according to the relation between cuff size and upper-arm circumference in critically ill patients. Crit Care Med 2000;28(2):371–376. DOI: 10.1097/00003246-200002000-00014.
Park MK, Menard SW, Yuan C. Comparison of auscultatory and oscillometric blood pressures. Arch Pediatr Adolesc Med 2001;155(1):50–53. DOI: 10.1001/archpedi.155.1.50.
Ghaffari S, Malaki M, Rezaeifar A, Abdollahi Fakhim S. Effect of peripheral edema on oscillometric blood pressure measurement. J Cardiovasc Thorac Res 2014;6(4):217–221. DOI: 10.15171/jcvtr.2014.015.
Kaur J, Bhargava S, Pooni P, Bhat D, Dhooria G, Arora K, et al. Comparison of non-invasive oscillometric and intra-arterial blood pressure measurements in children admitted to the pediatric intensive care unit. J Pediatr Intensive Care 2021. DOI: 10.1055/s-0041-1739264. Accessed on: 23 February 2023.
Goodwin A, Mazwi ML, Somer J, Schwartz SM, McEwan A, Eytan D. Blood pressure in critically ill children: Exploratory analyses of concurrent invasive and noninvasive measurements. Crit Care Explor 2021;3(12):e0586. DOI: 10.1097/CCE.0000000000000586.
Kukreti V, Mohseni-Bod H, Drake J. Management of raised intracranial pressure in children with traumatic brain injury. J Pediatr Neurosci 2014;9(3):207–215. DOI: 10.4103/1817-1745.147572.
Kumar R, Singhi S, Singhi P, Jayashree M, Bansal A, Bhatti A. Randomized controlled trial comparing cerebral perfusion pressure-targeted therapy versus intracranial pressure-targeted therapy for raised intracranial pressure due to acute CNS infections in children. Crit Care Med 2014;42(8):1775–1787. DOI: 10.1097/CCM.0000000000000298.
Sankar J, Singh M, Kumar K, Sankar MJ, Kabra SK, Lodha R. ‘Intermittent’ versus ‘continuous’ ScvO2 monitoring in children with septic shock: A randomised, non-inferiority trial. Intensive Care Med 2020;46(1):82–92. DOI: 10.1007/s00134-019-05858-w.
Nazir M, Wani W, Dar SA, Mir I-H, Charoo BA, Ahmad QI, et al. Lactate clearance prognosticates outcome in pediatric septic shock during first 24 h of intensive care unit admission. J Intensive Care Soc 2019;20(4):290–298. DOI: 10.1177/1751143719855202.
Ghosh S, Baranwal AK, Bhatia P, Nallasamy K. Suspecting hyperferritinemic sepsis in iron-deficient population: Do we need a lower plasma ferritin threshold? Pediatr Crit Care Med 2018;19(7):e367–e373. DOI: 10.1097/PCC.0000000000001584.
Choudhary R, Sitaraman S, Choudhary A. Lactate clearance as the predictor of outcome in pediatric septic shock. J Emerg Trauma Shock 2017;10(2):55–59. DOI: 10.4103/JETS.JETS_103_16.
Poddar B, Gurjar M, Singh S, Aggarwal A, Baronia A. Reduction in procalcitonin level and outcome in critically ill children with severe sepsis/septic shock-A pilot study. J Crit Care 2016;36:230–233. DOI: 10.1016/j.jcrc.2016.07.022.
Rudd KE, Kissoon N, Limmathurotsakul D, Bory S, Mutahunga B, Seymour CW, et al. The global burden of sepsis: Barriers and potential solutions. Crit Care 2018;22(1):232. DOI: 10.1186/s13054-018-2157-z.
Ravikumar N, Baranwal AK. Fluid bolus in hypotensive septic shock: Need to encourage critical care interventions outside the formal PICU. Pediatr Crit Care Med 2020;21(9):856–857. DOI: 10.1097/PCC.0000000000002420.
Parajuli B, Baranwal AK, Kumar-M P, Jayashree M, Takia L. Twenty-four-hour pretreatment with low dose (0.25 mg/kg/dose) versus high dose (0.5 mg/kg/dose) dexamethasone in reducing the risk of postextubation airway obstruction in children: A randomized open-label noninferiority trial. Pediatr Pulmonol 2021;56(7):2292–2301. DOI: 10.1002/ppul.25388.
Samprathi M, Baranwal AK, Gupta PK, Jayashree M. Pre-extubation ultrasonographic measurement of intracricoid peritubal free space: A pilot study to predict post-extubation airway obstruction in children. Int J Pediatr Otorhinolaryngol. 2020; 138: 110348.
Baranwal AK, Meena JP, Singhi SC, Muralidharan J. Dexamethasone pretreatment for 24 h versus 6 h for prevention of postextubation airway obstruction in children: A randomized double-blind trial. Intensive Care Med 2014;40(9):1285–1294. DOI: 10.1007/s00134-014-3358-9.
Chiolero A, Paradis G, Lambert M. Accuracy of oscillometric devices in children and adults. Blood Press 2010;19(4):254–259. DOI: 10.3109/08037051003606439.
Chandrasekhar A, Yavarimanesh M, Hahn JO, Sung SH, Chen CH, Cheng HM, et al. Formulas to explain popular oscillometric blood pressure estimation algorithms. Front Physiol 2019;10:1415. DOI: 10.3389/fphys.2019.01415.
O'Brien E, Petrie J, Littler W, de Swiet M, Padfield PL, O'Malley K, et al. The British hypertension society protocol for the evaluation of automated and semi-automated blood pressure measuring devices with special reference to ambulatory systems. J Hypertens 1990;8(7):607–619. DOI: 10.1097/00004872-199007000-00004.